Doubtful it would work. Anything is possible, but this approach would be my last resort. My first option would be straight exhaust.
1) Charcoal filters primarily useful on organics.
2) Acids require activated filters which are tricky.
3) Give-back of filters when turned off..
4) Wide divergence on quality and support of companies making this type of filtered hood…some good many bad!
Very Truly Yours,
Dr. Bob Haugen
Director of Product and Technology Development
Flow Sciences Inc.
2025 Mercantile Drive
Leland, NC 28451
Phone 910 332 4878
From: DCHAS-L Discussion List [mailto:dchas-l**At_Symbol_Here**MED.CORNELL.EDU]
On Behalf Of melissa.ballard**At_Symbol_Here**US.MICHELIN.COM
Sent: Monday, January 26, 2015 6:38 AM
Subject: [DCHAS-L] ductless fume hoods for acid digestion
Good morning all! I have to take a minute to get some additional opinions on a topic. I've been approached to provide my 2 cents on the appropriateness of using a ductless
fume hood for acid digestion (preparing samples for analysis via AA). This would be for use in an industrial quality lab.
I have my opinion; what is yours?
Melissa D. Ballard, MSPH, CIH, CHO
Industrial Hygienist / Industrial Hygiene Chemist
Michelin North America
1401 Antioch Church Road
Greenville, SC 29605
Previous post | Top of Page | Next post